Contemporary anesthesia ventilators incur a significant "oxygen cost"
Cast your vote
You can rate an item by clicking the amount of stars they wish to award to this item.
When enough users have cast their vote on this item, the average rating will also be shown.
Your vote was cast
Thank you for your feedback
Thank you for your feedback
MetadataShow full item record
Other TitlesLes ventilateurs d’anesthésie actuels exposent à une importante «dépense d’oxygène»
CitationCan J Anaesth 2004, 51(6):616-20
AbstractPURPOSE: Anesthesia ventilators use oxygen or oxygen/air mixtures to drive the bellows during controlled ventilation. As a practitioner may find himself in a situation that the only available oxygen source is a compressed oxygen cylinder, it is important to know the oxygen consumption of anesthesia ventilators during controlled ventilation. METHODS: We tested the Datex-Ohmeda 7900 ventilator mounted on an Excel 210 anesthesia machine under a variety of conditions. For comparison, we also tested the Ohmeda 7800 and the Dräger AV-2 ventilator under control conditions. All experiments were performed using a test lung. RESULTS: The oxygen consumption of the AV-2 and the Datex-Ohmeda ventilators averaged 302 +/- 17 L x hr(-1) and 564 +/- 68 to 599 +/- 56 L x hr(-1), respectively (P < 0.01 AV-2 vs 7800 and 7900). When using an E-type cylinder, this would result in a mean time to alarm of 93 min and 54 to 57 min, respectively. Decreased lung compliance increased the oxygen consumption to 848 +/- 16 L x hr(-1). CONCLUSIONS: Machine-driven mechanical ventilation incurs a significant "oxygen cost." We show that the amount of oxygen consumed by mechanical ventilation with contemporary anesthesia ventilators is influenced by patient-dependent factors and may greatly exceed the amount of oxygen delivered to the patient.
DescriptionTo access publisher full text version of this article. Please click on the hyperlink in Additional Links field
- Do new anesthesia ventilators deliver small tidal volumes accurately during volume-controlled ventilation?
- Authors: Bachiller PR, McDonough JM, Feldman JM
- Issue date: 2008 May
- Effect of inspiratory time on tidal volume delivery in anesthesia and intensive care unit ventilators operating in pressure control mode.
- Authors: Tung A, Drum ML, Morgan S
- Issue date: 2005 Feb
- [Mechanical ventilation in an anesthetic circle system using the lowest tidal volume--studies of 3 anesthesia ventilators in a lung model and an animal experiment].
- Authors: Fösel T, Schirmer U, Wick C, Pfenninger E, Siegler W, Altemeyer KH
- Issue date: 1991 Nov
- Improved flow and pressure capabilities of the Datex-Ohmeda SmartVent anesthesia ventilator.
- Authors: Katz JA, Kallet RH, Alonso JA, Marks JD
- Issue date: 2000 Feb
- Oxygen consumption with mechanical ventilation in a field anesthesia machine.
- Authors: Szpisjak DF, Lamb CL, Klions KD
- Issue date: 2005 Jun