Cast your vote
You can rate an item by clicking the amount of stars they wish to award to this item.
When enough users have cast their vote on this item, the average rating will also be shown.
Your vote was cast
Thank you for your feedback
Thank you for your feedback
AuthorsHardarson, Sveinn Hakon
Karlsson, Robert Arnar
Halldorsson, Gisli Hreinn
Zoega, Gunnar Már
Benediktsson, Jon Atli
Jensen, Peter Koch
MetadataShow full item record
CitationInvest. Ophthalmol. Vis. Sci. 2006, 47(11):5011-6
AbstractPURPOSE: To measure hemoglobin oxygen saturation (SO(2)) in retinal vessels and to test the reproducibility and sensitivity of an automatic spectrophotometric oximeter. METHODS: Specialized software automatically identifies the retinal blood vessels on fundus images, which are obtained with four different wavelengths of light. The software calculates optical density ratios (ODRs) for each vessel. The reproducibility was evaluated by analyzing five repeated measurements of the same vessels. A linear relationship between SO(2) and ODR was assumed and a linear model derived. After calibration, reproducibility and sensitivity were calculated in terms of SO(2). Systemic hyperoxia (n = 16) was induced in healthy volunteers by changing the O(2) concentration in inhaled air from 21% to 100%. RESULTS: The automatic software enhanced reproducibility, and the mean SD for repeated measurements was 3.7% for arterioles and 5.3% venules, in terms of percentage of SO(2) (five repeats, 10 individuals). The model derived for calibration was SO(2) = 125 - 142 . ODR. The arterial SO(2) measured 96% +/- 9% (mean +/- SD) during normoxia and 101% +/- 8% during hyperoxia (n = 16). The difference between normoxia and hyperoxia was significant (P = 0.0027, paired t-test). Corresponding numbers for venules were 55% +/- 14% and 78% +/- 15% (P < 0.0001). SO(2) is displayed as a pseudocolor map drawn on fundus images. CONCLUSIONS: The retinal oximeter is reliable, easy to use, and sensitive to changes in SO(2) when concentration of O(2) in inhaled air is changed.
DescriptionTo access publisher full text version of this article. Please click on the hyperlink in Additional Links field
- Choroidal oximetry with a noninvasive spectrophotometric oximeter.
- Authors: Kristjansdottir JV, Hardarson SH, Harvey AR, Olafsdottir OB, Eliasdottir TS, Stefánsson E
- Issue date: 2013 May 7
- Retinal oximetry.
- Authors: Hardarson SH
- Issue date: 2013 Mar
- Retinal vessel oximetry-calibration, compensation for vessel diameter and fundus pigmentation, and reproducibility.
- Authors: Hammer M, Vilser W, Riemer T, Schweitzer D
- Issue date: 2008 Sep-Oct
- Hemoglobin oxygen saturation (So2) in the human ocular fundus measured by reflectance oximetry: preliminary data in retinal veins.
- Authors: Crittin M, Schmidt H, Riva CE
- Issue date: 2002 Apr
- Oxygen saturation in human retinal vessels is higher in dark than in light.
- Authors: Hardarson SH, Basit S, Jonsdottir TE, Eysteinsson T, Halldorsson GH, Karlsson RA, Beach JM, Benediktsson JA, Stefansson E
- Issue date: 2009 May